General Description. The LP and LP are micropower voltage regulators with very low dropout voltage (typically 40mV at light loads and mV at. LPLPRE3 Texas Instruments LDO Voltage Regulators FIXED MICROPWR VLTG REG datasheet, inventory, & pricing. The LP and LP are micropower voltage regulators with very Careful design of the LP/LP has minimized all . See MIL/AERO Datasheet.
|Country:||Moldova, Republic of|
|Published (Last):||1 August 2009|
|PDF File Size:||1.11 Mb|
|ePub File Size:||18.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
We would probably buy a few, especially if the footprint was compact.
Fortunately, I disconnected it quickly enough to prevent any obvious permanent damage to the regulator. There is one error dayasheet the published circuit: For one low-power project, I have the possibility of powering the board from either a 9V solar cell or a 4. My power supply current-limited at about 1 amp.
If only the ATtiny26 had a bit more memory… Thanks, Jim. Shop Blog Forum Support. I especially like the plated through holes, which are fantastic for prototyping if you have to remount components several times. I could post the entire project if you think it would help sales of the ATtiny26 board I do intend to buy a few more — the sale price is very attractive, so be sure and save me a couple!
If I connect the 4. The program currently occupies out of bytes of program memory. The rainbow cable leading off screen goes to the two pushbuttons and the LCD.
National Semiconductor – datasheet pdf
I just tried taking a regulator part not on a board and applied from 4. Well, I connected a 4. Since there is some interest, my Geiger counter is based on a circuit described by Tom Napier in Circuit CellarNovember Can you try doing the regulator test again?
What I like about the ATtiny26 board is the convenient arrangement of two ports with access to power and ground at several points, coupled with the well-laid out prototyping area. The unit is currently monitoring the cosmic background radiation in my office 95 counts in seconds. Of course we had these boards made kp2950 a batch, so now each time Ratasheet populate one I have to bypass the power switch with tweezers to to clear that fuse. I had to jump! The HV generator is top right, leading to the connector for the V tube bias red arrow.
The JTAG fuse comes set, which makes some pins start up set high. The counter input INT0 is marked with a green arrow. There is a small current draw increase at 5. Thanks for checking on the regulator — maybe I shorted the power supply while hooking things up. I can always use diodes to switch between solar and backup supply, but I thought I could float charge the battery without such trickery.
Where did you get your Geiger tube? I suspect that no long term damage should result from this misuse, as long as one does not exceed other component ratings. Not realizing this we designed our main board which uses the pushbutton power switch with one of these connected to the OFF pin. I do that sort of thing all the time, and I have had no problems. See below for the exterior appearance. By the way, there are a few manufacturers of the part, and your board probably has the one from On Semiconductor.
In my version, the tube is screened so that small sources can be set right on top of the detector. Is the RS option useful? I expect that the regulator can tolerate this but it might add some unnecessary current drain.
The LP lp250 sheet does not give any insight into this situation. My most recent project with the ATtiny26 board is a Geiger counter. I could post the entire project if you think it would help sales of the ATtiny26 board. Both have a thin and very delicate metalized membrane for measurement of alpha radiation. Can you tell me what you like about it?
Datasheets search archive of electronic components datasheets
Did you look at the datasheet from that company? I do intend to buy a few more — the sale price is very attractive, so be sure and darasheet me a couple! By the way, I really like the board design.