The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Agilent A. 20 MHz Function /. Arbitrary Waveform Generator. Service Guide . Publication Number (order as manual set). Agilent A. Function/Arbitrary Waveform Generator .. trigger from the rear- panel Trig In connector, a manual trigger from the key, or a software (bus) trigger .
|Published (Last):||17 January 2014|
|PDF File Size:||19.55 Mb|
|ePub File Size:||9.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
If you set the carrier to a value less than the deviation frequency with FM enabledthe function generator will automatically adjust the deviation to the maximum value allowed with the present carrier frequency.
33220A Function / Arbitrary Waveform Generator, 20 MHz
Printed hardcopy manuals are available as an extra cost option. Index amplitude limitations 56 period modulating frequency frequency limitations 56 burst mode modulating source 88, modulation modes allowed front-panel selection 17 modulating waveform 87 pulse period limitations pulse waveform 70 modulation 85 3320a impedance period, pulse modulation waveshape affect on amplitude The function generator takes the specified number of points and expands them to fill waveform memory.
Arbitrary Waveforms 1 a dc voltage and 65, 64K data points. The security code is stored in non-volatile memory. For example, the following command sets the output frequency to 5 kHz: Page Index gated burstdefault address 48 invert waveform 67 gated burst mode front-panel configuration 48 IP address 50, gateway address service request dot notation 52 agilwnt noise setting address 48 further information 52 GPIB immediate trigger, address 48, The locations are numbered 0 through 4.
This is because the waveform may have a non-zero average value that would be added to the DC Offset. Burst Chapter 7 Tutorial Burst connector can be configured as an output to enable the A to trigger other instruments at the same time as its internal trigger occurs.
Since elliptical filters exhibit severe ringing for waveforms other than continuous sine waves, a linear-phase filter is used for all other waveform functions. IP address to a device on a network. The default is Linear. Chapter 1 Quick Start Quick Start Chapter 5 Error Messages Command Errors String data not allowed A character string was received but is not allowed for this command. For a linear sweep, the function generator varies the output frequency in a linear fashion during the sweep.
Page Tutorial In order to achieve the best performance from the Agilent A, it may be helpful for you to gain a better understanding of the internal operations of the instrument.
It does, however, give you an overview of the front- panel menus and many front-panel operations. The default is VPP. You can use burst in one of two modes as described below.
Chapter 3 Features and Functions Triggering External Triggering In the external trigger mode, the agillent generator will accept a hardware trigger applied to the rear-panel Trig In connector.
The agilebt arb name is the arbitrary waveform currently active selected with FUNC: Self-test failed; waveform filter path select relay to The default is agient degrees. Page Chapter 7 Tutorial Direct Digital Synthesis The A represents amplitude values by 16, discrete voltage levels or bit vertical resolution.
The default is 10 ms. The isolation between chassis ground circuits and floating circuits is controlled by an optical isolation barrier and a serial link. Page Chapter 3 Features and Functions Triggering External Triggering In the external trigger mode, the function generator will accept a hardware trigger applied to the rear-panel Trig In connector.
Index softkey labels 4 questionable data register trigger output software bus trigger,register diagram trigger source standard event register sweeps software revision status byte register Sync 3322a0 software, connectivity 15 status system symmetry 65 sound beeper Page 84 Chapter 3 Features and Functions Frequency Modulation FM Frequency Deviation The frequency deviation setting represents the peak variation in frequency of agileng modulated waveform from the carrier frequency. Then, rotate the handle to the desired position.
Press and then select the Store State or Recall State softkey. Select the byte order for binary transfers in the block mode using the DATA: Arbitrary Waveform Commands Note.
You must insert a blank space 332200a separate a parameter from a command agileng. Page Index softkey labels 4 questionable data register trigger output software bus trigger,register diagram trigger source standard event register sweeps software revision status byte register Sync signal software, connectivity 15 status system symmetry 65 sound beeper Chapter 7 Tutorial Direct Digital Synthesis The A represents amplitude values by 16, discrete voltage levels or bit vertical resolution.
Agilent Technologies 33220A User Manual
Page This connector accepts TTL-compatible levels and is referenced to chassis ground not floating ground. The following steps show you how to configure a ms pulse waveform with a pulse width of 10 ms and edge times of 50 ns. Press the Create New softkey to start the waveform editor. Instrument State Storage The function generator has five storage locations in non-volatile memory to store instrument states. The default is 10 Hz. You can create an arbitrary waveform from the front panel as described in the following section, or you can use the Agilent IntuiLink software provided on the CD-ROM included with the Agilent A.
Index If you have questions relating to the operation of the Agilent A, call in the United States, or contact your nearest Agilent Technologies Office. Page Chapter 3 Features and Functions System-Related Operations Beeper Control Normally, the function generator will emit a tone when an error is generated from the front-panel or over the remote interface. The variation in frequency of the modulated waveform from the carrier frequency is called the frequency deviation. The CD auto-starts and provides a Welcome page with instructions.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
The default offset is 0 volts for all functions. Select linear or logarithmic spacing for the sweep. To rack mount two instruments side-by-side, order lock-link kit and flange kit Unsecure or secure the instrument for calibration. Select whether the function generator uses true-high or true-low logic levels on the rear-panel 33220s In connector for an externally-gated burst.
Status Byte Register Commands See the table on page for the register bit definitions.